Scope of Application: Can measure the thickness of the non magnetic coating layer on magnetic metal substrates such as steel, and the thickness of the non conductive coating layer on non magnetic metal substrates such as copper and aluminum. Automatic Identification: The iron and aluminum dual purpose probe can automatically identify iron, aluminum and non metallic substrates, and can quickly and automatically switch. Fast and Simple Measurement: The measurement is fast, and a measurement is completed in 0.5 seconds. No need to calibrate, just adjustment, the only key operation is extremely simple. Rubine Probe and Measurement Mode: The abrasion resistance of the rubine probe ensures the long term effective use of the instrument."Fe", "NFe" and "Fe/NFe" three measurement modes can be set. Stable Measurement: Even temperature changes will not affect the measurement, and the reading remains stable to ensure very good repeatability throughout the measurement process. Item Type: Paint Coating Thickness Gauge Model: LS230 Material: ABS Color: As Pictures Shown Weight: Approx.
247g / 8.7oz Battery: 2 x AAA 1.5V(Not Included) Application: Paint Film, Coating Thickness Measurement. Fe: Hall Effect / NFe: Eddy Current.0.01mm: (1.00mm-2.00mm). Convex : 5mm / 0.2in Concave : 25mm / 1.0in. Fe: 0.2mm / NFe: 0.05mm. 128x64 Dot Matrix Organic Light Emitting Diode. 2pcs 1.5V AAA Alkaline Batteries. 101x62x28mm / 4.0x2.4x1.1in. On/Off Power on: Short press the button on the instrument to power on. After power on, the version number and serial number will be displayed, and then the last recorded historical data interface will be entered. Shutdown: Long press the button on the instrument to shut down; or automatically shut down after 3 minutes without operation. Instrument settings In the shutdown state, press and hold the key for 3 seconds to enter the instrument setting interface. After entering the setting interface, if there is no key operation for more than 20 seconds, the instrument will automatically shut down. The short key is the "select" function; the long key longer than 3 seconds and less than 5 seconds is the "confirm" function; the long key (more than 5 seconds is not released), the instrument exits the setting and shuts down, and the setting is invalid. 2.1 Measurement matrix selection There are three measurement modes: iron based measurement (Fe), non-ferrous-based measurement (NFe) and automatic identification (Fe/NFe). Under normal circumstances, Fe/NFe can be used to automatically recognize the mode. When the measurement matrix has been confirmed, you can select Fe or NFe as the fixed measurement mode. Press the short button to select the substrate to be measured, and press the button for 3 seconds to confirm the substrate selection and enter the measurement state. 2.2 Language settings The instrument has two displays in Chinese and English, and the factory default is Chinese display. Setting method: Press and hold the key in the shutdown state to enter the setting interface, short press key to select "Language", long press key for 3 seconds to enter the Chinese and English selection interface, then short press key to select the desired language, long press key for 3 seconds, language setting Finish, enter the measurement interface. 2.3 Unit settings The instrument can be set to or imperial units, and the factory default is the unit um. Setting method: In the shutdown state, long press the key to enter the setting interface, short press key to select "Unit", long press key for 3 seconds to enter the unit selection interface, then short press key to select the desired unit, long press key for 3 seconds, the unit setting is completed To enter the measurement interface. Adjustment When the instrument is used for the first time, after the battery is changed, the measuring material is changed or the ambient temperature changes, in order to reduce the measurement error, the adjustment operation should be performed. It is strongly recommended to use the same kind of uncoated workpiece for adjustment (because there may be differences in physical properties such as magnetism or conductivity between the tested material and the adjustment board). If there is no uncoated workpiece, you can use the attached aluminum adjusting board or iron adjusting board to adjust the. Please select the correct one according to the different materials to be measured.
3.1 Use the instrument to measure the adjusting board or uncoated workpiece (substrate), and the instrument will display a measurement value (when measuring, press the probe of the instrument vertically on the center of the adjusting board or the substrate, and keep the probe stable without tilting or shaking). 3.2 Keep the probe still, press and hold the button for 3 seconds, the instrument will prompt "Please press the probe tightly for adjustment". 3.3 After hearing the buzzer prompt, the instrument prompts "please lift the probe more than 15cm", release the button, lift the probe and leave the adjustment board or the substrate more than 15cm. 3.4 After hearing the buzzer sound again, the data of 0.0 is displayed, and the adjustment is completed. 3.5 After the adjustment is completed, place the test piece marked with the standard value on the adjustment board or the substrate for measurement.
If the value is stable and consistent with the value calibrated on the standard piece (the error is within ±5um), the instrument can be used normally. Note: Due to the roughness of the workpiece, dust, scratches, etc. After adjustment, it may not necessarily display 0µm when measuring the same position again. The operation of the instrument must be correct and skilled, otherwise it will also cause the instability of the measured value. Measure (1) Hold the position of the anti skid grooves on the lower part of the instrument with your fingers.
(2) Press the probe of the instrument vertically on the of the object to be measured, keep the probe stable, do not tilt or shake, the measurement result will appear on the screen with a buzzer prompt. (3) To continue the measurement, you can lift the probe away from the object to be measured, and then follow step (2) again.
View measurement records In measurement mode, press the short key to view historical data. The instrument stores 9 sets of data. When 9 sets of data are exceeded, the oldest recorded value is automatically deleted. 1 is the most recent test data, and the recorded data will not be lost when shutting down. 1 x Coating Thickness Gauge(Battery Not Included).
It is strictly forbidden to slide the probe on the of the car, which will cause damage to the paint and the instrument. Please ensure that the paint of the car is clean.Dust and dirt on the paint will affect the accuracy of the measurement. When the instrument displays "low battery", a new battery needs to be replaced. Our listing price includes the value of goods, packaging cost, and export. Please check with your country's. We will reply your message within 48 hours. If you do not receive our reply within 48 hours, pls. Post a message again because sometimes our message system may block some emails. If you are satisfied with our service, pls. Our aim is to provide Level Customer Service, your recognition will encourage us greatly and serve you better. We strive to keep all our customers satisfy. We will do our up most to exceed your expectations in customer service, communication and dispatching of your orders. This item is in the category "Business, Office & Industrial\Test, Measurement & Inspection\Test Meters & Detectors\Other Test Meters & Detectors".
The seller is "binggoshop" and is located in this country: CN. This item can be shipped worldwide.